For Healthcare Professionals OnlyPartner Portal
For questions about coronavirus/COVID-19 and using NIOX VERO®, please visit the NIOX VERO Coronavirus/COVID-19 FAQ.
NIOX VERO® is intended for use in the electromagnetic environment specified below. The user of NIOX VERO should assure that it is used in such an environment.
|RF emissions - CISPR 11||Group 1|
|RF emissions - CISPR 11||Class A|
|Harmonic emissions - IEC 61000-3-2||Class D|
|Voltage fluctuations / flicker emissions - IEC 61000-3-3||Complies|
NIOX VERO uses RF energy only for its internal function. Therefore, its RF emissions are very low and are not likely to cause any interference in nearby electronic equipment.
NIOX VERO is suitable for use in all establishments other than domestic, and may be used in domestic establishments and those directly connected to the public low-voltage power supply network that supplies buildings used for domestic purposes, provided the following warning is heeded:
Warning: This equipment/system is intended for use by healthcare professionals only. This equipment/system may cause radio interference or may be necessary to take mitigation measures, such as re-orienting or relocating NIOX VERO or shielding the location.
RF: Radio Frequency