NIOX VERO® Products Coronavirus FAQ
COVID-19: A personal message to our customers and partners
The world is facing an unprecedented human challenge with COVID-19. We stand together in supporting healthcare professionals around the world, who we have worked in close partnership with for many years.
Recognizing the extreme challenges that healthcare professionals face every working day, we wanted you to know that we are here for you and that the entire NIOX® team are available to support you during these difficult times. Our Medical Affairs team is ready and waiting to answer any questions or to provide support on FeNO testing.
Behind the scenes, the NIOX team have been working tirelessly to ensure that we have been able to safely maintain supply of NIOX devices and accessories, so that healthcare professionals can deliver the care that their asthma patients need at this critical time. It is important to mention that the health and wellbeing of our people is of the upmost importance. To maintain the reliable supply of NIOX, we have devised new working practices and safety measures to ensure that the team can work safely and effectively.
We are very proud of the Circassia team for their dedication and passion during these challenging times and want to take this opportunity to publicly thank them.
We know that science will win, in the end. In the meantime, you can depend on the Circassia team to stand with you and support you.
Ian Johnson (Executive Chairman)
Jonathan Emms (Chief Operating Officer)
What have we done so far?
During these unprecedented times, we have taken steps to introduce procedures that keep our colleagues safe and ensure our customers are fully supported.
Currently, all Circassia colleagues who can do so are working from home, but you can still contact us through your local representative, head office, customer service, technical support and operations by using the normal telephone number or email address.
As part of our commitment, we have implemented virtual training sessions and a remote installation service to support our customers and help assist with the diagnosis and management of patients with asthma, using FeNO testing.
FeNO testing with NIOX VERO
How can I continue to use FeNO to manage my patients with asthma during this period?
It is recommended that maintaining asthma control and continuing to follow the individuals’ Asthma Plan is especially critical during this pandemic of the respiratory disease COVID-191,2,3
The NIOX VERO device utilizes single-use patient mouth-filters, designed to prevent the spread of pathogens between patients and every NIOX sensor pack includes sufficient patient filters to accommodate a new filter for each test.4 Viral and bacterial filter efficiency is answered below.
Local infection control guidelines and policies should always be followed in the first instance. In addition, the World Health Organization published helpful guidance to provide further advice on risk management following suspected coronavirus infection.5,6 Based on recent government advice, we would endorse the following strategies to support the use of FeNO testing:7
- Where possible, FeNO testing should be carried out in a single room with the doors shut
- Full PPE should be worn according to local guidelines
- Disposable gloves to be worn by each patient when they are holding the device handle
- Appropriate hand hygiene measures throughout
How effective are the NIOX VERO patient filters?
The bacterial and viral filtration efficiencies of the disposable NIOX VERO mouth filter are exceptionally high (>99.97%8,9)
The disposable, single-use mouthpiece filter must be changed for each new patient but may be used on multiple occasions in the same session (by the same patient).4
NIOX VERO mouth filters meet international standards of filtration performance for, “Breathing System Filters for Anaesthetic and Respiratory Use” (ISO 23328-1, EN 13328-1), which many other respiratory devices in hospital settings also adhere to. The filters have been tested for bacterial filtration efficiency (BFE) and viral filtration efficiency (VFE) to the highest ASTM standard F2101. The BFE of the filter is 99.99%8 and the VFE is 99.97%.9
Several in vitro studies were independently conducted to test the viral efficiency of the NIOX mouthpiece filter. A microbe 4 times smaller than the size of SARS-CoV-2 (COVID-19 virus) was prevented from passing through NIOX filter with 99.97% efficiency.9-12
How should I clean and disinfect the NIOX VERO?
NIOX VERO should be regularly cleaned and decontaminated, as with all surfaces in a patient-care setting.
Alcohol containing disinfectants (such as ethanol and isopropyl alcohol) must not be used for cleaning the NIOX VERO device or breathing handle, as this will affect both the performance and results generated.4 However, several products are compatible with NIOX devices including:
- Sodium hypochlorite (<0.65%),
- Quaternary ammonium compounds such as Benzalkonium chloride (<5%),
- Biguanides such as Polyhexamethylene biguanide, Chlorhexidine digluconate or Poly (hexamethylene biguanide) hydrochloride (<1%),
- Isothiazolinones such as 2-methyl-2H-isothiazol-3-one or 5-chloro-2-methyl-2H-isothiazol-3-one (<0.01%).
In regards to COVID-19 specifically: Sodium hypochlorite 0.5-0.65% is effective against SARS-CoV-2, (the coronavirus causing COVID-19) and can be used to clean the NIOX VERO device itself.13-18 Benzalkonium chloride 0.05-0.2% is also an effective cleaning agent against SARS-CoV-2 and is compatible with NIOX.13-18
For any further questions
If you would like more information or have further questions, please contact our Medical Affairs team at email@example.com.